Force, Stiffness and Viscous Damping Control of a Stewart-Platform-Type Ankle-Foot Rehabilitation Assist Device with Pneumatic Actuator
Takayuki Onodera*, Eiji Suzuki*, Ming Ding**,
Hiroshi Takemura*, and Hiroshi Mizoguchi*
*Tokyo University of Science, 2641 Yamazaki, Noda, Chiba 278-8510, Japan
**RIKEN-TRI Collaboration Center for Human-Interactive Robot Research, Advanced Science Institute, RIKEN, 2271-130 Anagahora, Shimoshidami, Moriyama-ku, Nagoya 463-0003, Japan
The number of physically disabled people in need of rehabilitation is increasing. Unfortunately, there is a shortage of physical therapists specializing in such rehabilitation. This has increased the demand for rehabilitation assist devices that can lessen the burden of physical therapists. In this study, the authors develop a device that can assist in the rehabilitation of the ankle joint by employing a Stewart-platform (SP)-type parallel-link mechanism. With the SP-type parallel-link mechanism, it is possible to measure and control six degrees-of-freedom (DOFs) of ankle-foot movement during rehabilitation. Because the device enables the measurements of the ankle and foot, it is possible to calculate the instantaneous center of the ankle joint. In previous studies, the authors proposed methods to calculate and control the posture of the ankle and foot by an SP-type parallel-link mechanism and verified their accuracy. In this paper, the authors propose a method for force control using the device and also verify its accuracy. Using this device, the force acting on the ankle-foot can be controlled by six air cylinders. The force produced by a single air cylinder is determined by controlling the pressures in the extension and retraction directions. The accuracy of the force control method is verified for a single air cylinder and for the assist device when all six air cylinders are engaged. Results show that the accuracy of the single air cylinder has a mean square error of 0.24 N or less, while those for force control of the entire device are 4.2 N or less for parallel translation and 3.2 Nm or less for rotation. This demonstrates a satisfactory accuracy. To incorporate rehabilitation assistance by means of stiffness or viscous damping in the future, the authors also propose methods to control the coefficients of stiffness and viscous damping of the air cylinder and verify their accuracy. The mean square errors for the accuracies in controlling the coefficients of stiffness and viscous damping are 3.4 N/m and 1.4 Ns/m, respectively, thus verifying the validity of the proposed methods.
Hiroshi Takemura, and Hiroshi Mizoguchi, “Force, Stiffness and Viscous Damping Control of a Stewart-Platform-Type Ankle-Foot Rehabilitation Assist Device with Pneumatic Actuator,” J. Robot. Mechatron., Vol.25, No.6, pp. 897-905, 2013.
-  M. Ding, J. Ueda, and T. Ogasawara, “Pinpointed Muscle Force Control Using a Power-assisting Device,” J. of the Robotics Society of Japan, Vol.27, No.9, pp. 1037-1045, 2009.
-  R. J. Farris, H. A. Quintero, T. J. Withrow, and M. Goldfarb, “Design and Simulation of a Joint-Coupled Orthosis for Regulating FES-Aided Gait,” 2009 IEEE Int. Conf. on Robotics and Automation (ICRA 2009), Kobe Int. Conf. Center, Kobe, Japan, pp. 1916-1922, May 12-17, 2009.
-  H. I. Krebs, J. Celestino.D. Williams, M. Ferraro, B. Volpe, and N. Hogan, “A wrist extension for MIT-MANUS,” Advances in Rehabilitation Robotics, pp. 377-390, 2004.
-  S. Lee and Y. Sankai, “Power assist control for walking aid with hal-3 based on EMG and impedance adjustment around knee joint,” IEEE RSJ Int. Conf. Intelligent Robots and Systems, pp. 1499-1504, 2002.
-  A. Roy, H. I. Krebs, S. L. Patterson, T. N. Judkins, I. Khanna, L.W. Forrester, R. M. Macko, and N. Hogan, “Measurement of human ankle stiffness using the anklebot,” The 2007 IEEE 10th Int. Conf. on Rehabilitation Robotics, pp. 358-363, June 12-15, 2007.
-  D. G. Wright, S. M. Desai, and W. H. Henderson, “Action of the subtalar and ankle-joint complex during the stance phase of walking,” J. of Bone and Joint Surgery, No.46, pp. 361-382, 1964.
-  A. Lundberg, O. K. Svensson, G. Nemeth, and G. Selvik, “The axis of rotation of the ankle joint,” The J. of Bone and Joint Surgery, No.71(B), pp. 94-99, 1989.
-  J. H. Hicks, “The mechanics of the foot: I. the joints,” J. of Anatomy, No.87, pp. 345-357, 1953.
-  M. Takaiwa, T. Noritsugu, N. Ito, and D. Sasaki, “Wrist rehabilitation device using pneumatic parallel Manipulator based on EMG signal,” Int. J. of Automation Technology, Vol.5, No.4, pp. 472-477, 2011.
-  D. Stewart, “A Platform with Six Degrees of Freedom,” Institution of Mechanical Engineers, Vol.180, No.15, pp. 371-386, 1966.
-  T. Onodera, M. Ding, H. Takemura, and H. Mizoguchi, “Posture Control Using New Ankle-Foot Assist Device with Stewart Platform Type Parallel Link Mechanisms,” 2012 IEEE Int. Conf. on Robotics and Biomimetics (ROBIO2012), Guangzhou, China, pp. 1385-1390, Dec. 10-14, 2012.
-  T. Onodera, M. Ding, H. Takemura, and H. Mizoguchi, “Design and Development of Stewart Platform-Type Assist Device For Ankle-Foot Rehabilitation,” The 2012 Int. Conf. on Innovative Engineering Systems (ICIES 2012), pp. 1-6, Dec. 7-9, 2012.
-  T. Ohtani, “Oral History in Instrument and Control Engineering – Dr. Toshiyuki Kitamori; I-PD Control and its Design Method,” J. of the Society of Instrument and Control Engineers, Vol.47, No.11, pp. 968-969, 2008.
-  S. Lee, J. Song,W. Choi, and D Hong, “Position control of a Stewart platform using inverse dynamics control with approximate dynamics,” Mechatronics, Vol.13, No.6, pp. 605-619, July 2003.
-  K. Kosuge, K. Takeo, T. Fukuda, H. Kitayama, N. Takeuchi, and H. Murakami, “Force control of parallel link manipulator with hydraulic actuators,” Proc. of the 1996 IEEE Intemational Conf. on Robotics and Automation Minneapolis, Vol.1, pp. 305-310, April 1996.
-  X. Shen and M. Goldfarb, “Simultaneous Force and Stiffness Control of a Pneumatic Actuator,” J. of Dynamic Systems, Measurement, and Control, Vol.129, No.4, pp. 425-434, 2007.
-  M. Ding, T. Onodera, R. Ikeura, H. Takemura, and H. Mizoguchi, “Position, Force and Stiffness Control of a Stewart-Platform-Type Ankle-Foot Assist Device,” 2012 ASME Dynamic Systems and Control Conf., Ft. Lauderdale, FL, Oct. 17-19, 2012.
- [a] Policy Planning Division, Department of Health and Welfare for Persons with Disabilities, Social Welfare and War Victims’ Relief Bureau, Ministry of health labour and welfare japan, “Survey on persons with physical disability,” Ministry of health labour and welfare Japan.
http://www.mhlw.go.jp/toukei/saikin/hw/shintai/06/dl/01_0001.pdf [Accessed January 15, 2013]
- [b] World Stroke Organization,
http://www.world-stroke.org [Accessed March 1, 2013]
Supporting Online Materials:
Copyright© 2013 by Fuji Technology Press Ltd. and Japan Society of Mechanical Engineers. All right reserved.