Drip Adjuster: Use of an LED Display to Manually Adjust Intravenous Fluid Infusion Rate
Takeshi Ando*,**, Noriyoshi Tanaka*,***, Kenji Yamada*,
and Yuko Ohno*
*Robotics and Design for Innovative Healthcare, Graduate School of Medicine, Osaka University, 1-7 Yamada-oka, Suita, Osaka 565-0871, Japan
**Graduate School of Creative Science and Engineering, Faculty of Science and Engineering, Waseda University, 59-309, 3-4-1 Ohkubo, Shinjuku, Tokyo 169-8555, Japan
***The Nursing Department, Tokai University Hospital, Isehara, Kanagawa, Japan
Errors in intravenous infusion rates are common at hospitals. A previous study found that actual infusion rates differ significantly from those instructed by doctors. The technique used to adjust the drip rate using a watch is a difficult skill to learn. In this article, we present our recently developed drip adjuster, which makes it easier for a nurse, for example, to adjust the drip rate using an LED display controlled by an Arduino microcontroller. We analyzed a high-speed video of falling infusion drops and imitated dripping by changing the brightness and positioning of light displayed by a row of five LEDs, enabling nurses to easily synchronize LED lighting with the growth and falling of droplets. We then evaluated the accuracy of the drip rate when six nurses used the drip adjuster versus using a watch. We found a significant difference in accuracy between the two methods, with a dramatic increase from 40% accuracy using a watch to 83% accuracy using the drip adjuster in the achievement of an accurate drip rate. The drip adjuster is a simple, effective device that can be used to assist in adjusting the drip rates of intravenous infusions.
and Yuko Ohno, “Drip Adjuster: Use of an LED Display to Manually Adjust Intravenous Fluid Infusion Rate,” J. Robot. Mechatron., Vol.24, No.3, pp. 452-457, 2012.
-  L. Portoni et al., “User-oriented views in health care information systems,” IEEE Trans. Biomed. Eng., Vol.49, No.12, pp. 1387-1398, Dec. 2002.
-  J. R. Slagle et al., “Monitor: an expert system that validates and interprets time-dependent partial data based on a cystic fibrosis home monitoring program,” IEEE Trans. Biomed. Eng., Vol.36, No.5, pp. 552-558, 1989.
-  M. D. Schwartz and D. Morton, “The Emerging Field of Clinical Engineering and Its Accomplishments,” IEEE Trans. Biomed. Eng., Vol.31, No.12, pp. 743-748, 1984.
-  J. H. Carrington et al., “Physical Arrangements at the Bedside in Support of Automated Systems for Patient Care,” IEEE Trans. Biomed. Eng., Vol.18, No.2, pp. 149-154, 1971.
-  G. J. Kost, “Preventing Medical Errors in Point-of-Care Testing,” Arch. Pathol. Lab. Med., Vol.125, No.10, pp. 1307-1315, 2001.
-  D. Simmons et al., “Tubing Misconnections: Normalization of Deviance,” Nutr. Clin. Pract., Vol.26, pp. 286-293, 2011.
-  R. Flin and R. Patey, “Improving patient safety through training in non-technical skills,” BMJ, Vol.339, b3595, Sep., 2009.
-  R. Patey, R. Flin, B. H. Cuthbertson, L. MacDonald, K. Mearns, J. Cleland, and D. Williams, “Patient safety: helping medical students understand error in healthcare,” Qual Saf Health Care, Vol.16, No.4, pp. 256-259, 2007.
-  J. Garbutt, D. R. Brownstein, E. J. Klein, A. Waterman, M. J. Krauss, E. K. Marcuse, E. Hazel, W. C. Dunagan, V. Fraser, and T. H. Gallagher, “Reporting and Disclosing Medical Errors: Pediatricians��� Attitudes and Behaviors,” Arch Pediatr Adolesc Med, Vol.161, No.2, pp. 161-179, 2007.
-  J.-H. Youna, H. Alia, H. Sharifb, and B. Chhetria, “RFID-based information system for preventing medical errors,” Mobile and Ubiquitous Systems: 6th Annual Int. Networking & Services, MobiQuitous, pp. 1-6, 2009.
-  L. Kohn, J. Corrigan, and M. Donaldson, “To Err Is Human: Building a Safer Health System,” Committee on Quality of Health Care in America, Institute of Medicine, 2000.
-  D. Morrow, R. North, and C. D. Wickens, “Reducing and Mitigating Human Error in Medicine,” Reviews of Human Factors and Ergonomics, pp. 254-296, 2005.
-  J. Gladstone, “Drug administration errors: a study into the factors underlying the occurrence and reporting of drug errors in a district general hospital,” J Adv Nurs, Vol.22, pp. 628-637, 1995.
-  P. Y. Han, I. D. Coombes, and B. Green, “Factors predictive of intravenous fluid administration errors in Australian surgical care wards,” Qual Saf Health Care, Vol.14, pp. 179-184, 2005.
-  J. C. Rooker and D. A. Gorard, “Errors of intravenous fluid infusion rates in medical inpatients,” Clin Med., Vol.7, No.5, pp. 482-485, Oct. 2007.
-  M. Husch, C. Sullivan, D. Rooney, C. Barnard, M. Fotis, J. Clarke, and G. Noskin, “Insights from the sharp end of intravenous medication errors: implications for infusion pump technology,” Qual Saf Health Care, Vol.14, No.2, pp. 80-86, 2005.
-  S. Paparella, ““Get smart” about infusion devices,” J. Emerg Nurs, Vol.35, No.1, pp. 52-54, 2008.
-  A. D. Harding, “Use of intravenous smart pumps for patient safety,” J Emerg Nurs, Vol.37, No.1, pp. 71-72, 2010.
-  L. J. Murdoch and V. L. Cameron, “Smart infusion technology: a minimum safety standard for intensive care?,” Br J Nurs, Vol.17, No.10, pp. 630-636, 2008.
-  L. Ragatz, Z.-Y. Jiang, C. E. Bauer, and H. Gest, “Macroscopic phototactic behavior of the purple photosynthetic bacterium,” Archives of Microbiology, Vol.163, pp. 1-6, 1995.
-  J. Sarik and I. Kymissis, “Lab kits using the Arduino prototyping platform,” 2010 IEEE Frontiers in Education Conf. (FIE), pp. 27-30, 2010.
-  Y. Kato, “Splish: A Visual Programming Environment for Arduino to Accelerate Physical Computing Experiences,” 2010 Eighth Int. Conf. on Creating Connecting and Collaborating through Computing (C5), pp. 3-10, 2010.
-  S. Negru, “A conceptual architecture of an Arduino-based socialemotional interactive system,” 2010 IEEE Int. Conf. on Intelligent Computer Communication and Processing (ICCP), pp. 93-98, 2010.
-  L. Buechley and M. Eisenberg, “The LilyPad Arduino: Toward Wearable Engineering for Everyone,” IEEE Pervasive Computing, Vol.7, No.2, pp. 12-15, 2008.
Copyright© 2012 by Fuji Technology Press Ltd. and Japan Society of Mechanical Engineers. All right reserved.