A Study on Optimal Voltage of Electromagnet for Precision Measuring Robot During Surface Roughness Measurement by Vibration Analysis
Kazuhide Tanaka, Dan Nakaya, Yuki Kondo, and Ichiro Yoshida
3-7-2 Kajinocho, Koganei-shi, Tokyo 184-8584, Japan
In electric power stations, precision surface roughness measurements are performed for environmental loading reduction, quality assurance, and safety. These measurements are performed manually at high places, narrow places, uncomfortable environments, etc. Therefore, workers in power stations experience a lot of hardship and are exposed to danger. To solve these issues, this study researched and developed a crawler-type robot with high measurement accuracy. Conventionally, robots that supply workpieces for surface roughness instruments have been developed. However, to the best of our knowledge, robotization and self-propelled precision measurement instruments have not been developed. Usually, a precision measurement instrument is designed for increased stiffness and stability because high measurement accuracy is the highest priority. However, if the stiffness and stability of the robot are as high as those of the precision measurement instrument, a problem occurs in the robot operation. Therefore, we propose a precision measurement unit using electromagnets and a crawler-type self-propelled robot to equip the unit. In a previous study, vibration analysis experiments using the impulse response method were performed on a precision measuring robot. In this study, the relationships between the voltages applied to the electromagnet and the reductions in the vibration magnitudes were determined by analyzing the vibrations of the robot during measurement. Furthermore, an optimal voltage of the electromagnets of the precision measuring robot to reduce vibrations was determined. From the results of the vibration analysis, the authors demonstrated that the optimal voltages were 9 and 12 V, and the precision measurement unit confirmed the effectiveness and validity of vibration reduction and improved measurement accuracy.
-  K. Onishi, Y. Kohata, N. Onishi, and K. Shinoyama, “Development of manipulators for nuclear power plants maintenance and an hierarchical open system,” J. of the Robotics Society of Japan, Vol.34, No.4, pp. 280-290, 2016.
-  S. Kim and C. Choi, “The robotic technology applications on nuclear power plants in Korea,” J. of the Robotics Society of Japan, Vol.23, No.5, pp. 548-551, 2005.
-  L. E. Parker and J. V. Draper, “Robotics applications in maintenance and repair,” S. Y. Nof (Ed.), “Handbook of Industrial Robotics, 2nd Edition,” Wiley, pp. 1-16, 1999.
-  I. Yoshida, K. Tanaka, and H. Kim, “Precision measurement robot for assisting efficiency improvement of energy plant,” Abstract of The Hyper-Interdisciplinary Conf. 2018, TEPIA Advanced Technology Gallery, Tokyo, Japan, Vol.7, 84, 2018 (in Japanese).
-  I. Yoshida and K. Tanaka, “Development of precision measuring robot of crawler type and feasibility study of measuring unit for improvement of measurement accuracy,” Trans. on GIGAKU, Vol.6, No.1, pp. 13-20, 2019.
-  JIS B 0601:2013, “Geometrical Product Specifications (GPS) – Surface texture: Profile method-Terms, definitions and surface texture parameters,” ISO 4287:2009, 2013.
-  ISO 16610-20:2015, “Geometrical product specifications (GPS) – Filtration – Part 20: Linear profile filters: Basic concepts,” 2015.
-  JIS B 0634:2017 (ISO 16610-21:2011), “Geometrical product specifications (GPS) – Filtration – Linear profile filters: Gaussian filters,” 2017.
-  ISO 16610-60:2015, “Geometrical product specification (GPS) – Filtration – Part 60: Linear areal filters – Basic concepts,” 2015.
-  JIS B 0635:2018 (ISO 16610-61:2015), “Geometrical product specifications (GPS) – Filtration – Linear areal filters – Gaussian filters,” 2018.
-  https://mect-japan.com/2017/visitor/information.php?ChargeNo=134 [Accessed April 7, 2020]
-  https://www.youtube.com/watch?v=ZI1KJfxUoWI [Accessed April 7, 2020]
-  https://www.qualitymag.com/articles/95352-robot-robots-everywhere [Accessed April 7, 2020]
-  K. Tanaka and I. Yoshida, “Validation of accuracy improvement effect and factor of precision measuring robot using electromagnet by vibration analysis,” Abstract of the 4th STI-GIGAKU (Int. Conf. on Science of Technology Innovation 2019), Nagaoka University of Technology, Niigata, Japan, Poster No.STI-7-7, 76, 2019.
-  K. Tanaka and I. Yoshida, “Validation of accuracy improvement effect and factor of precision measuring robot using electromagnet by vibration analysis,” Trans. on GIGAKU, Vol.7, No.1, pp. 07003-1-07003-7, 2020.
-  http://www.ait-ultrasonic.jp/details1122.html [Accessed April 7, 2020]
-  K. Tanaka and I. Yoshida, “A study of precision measuring robot,” Proc. of the Abrasive Technology Conf. 2019, Saitama University, Saitama, Japan, pp. 219-220, 2019 (in Japanese).
-  T. Umezu and D. Kono, “Machining Process for a Thin-Walled Workpiece Using On-Machine Measurement of the Workpiece Compliance,” Int. J. Automation Technol., Vol.13, No.5, pp. 631-638, 2019.
-  T. Moriwaki, K. Ashida, and K. Iwata, “Measurement of Dynamics of Machine Tool Structures by Shock Excitation Method (1st Report),” Precision Machine, Vol.44, No.6, pp. 724-729, 1978 (in Japanese).
-  T. Moriwaki and K. Iwata, “Measurement of Dynamics of Machine Tool Structures by Shock Excitation Method (2nd Report),” Precision Machine, Vol.45, No.522, pp. 848-853, 1979 (in Japanese).
-  https://svmeas.rion.co.jp/download/catalog/PV-97 [Accessed April 7, 2020]
-  Institute of Electrical Engineers of Japan, “Denkidenshikiki to Shindou Shougeki, Corona Publishing,” p. 9, 1988 (in Japanese).
This article is published under a Creative Commons Attribution-NoDerivatives 4.0 Internationa License.