Paper:

# Implementation of High-Precision Magnetostrictive-Type Liquid Level Measurement System UsingWavelet Transform

## Woo-Jin Choi^{*} and John-Tark Lee^{**,†}

^{*}HANLA IMS Co., Ltd., 115 Hwajeonsandan 1(il)-ro, Gangseo-gu, Busan, Korea

^{**}Department of Electrical Engineering, Dong-A University, 37 Nakdong-Daero 550beon-gil, Saha-gu, Busan, Korea

^{†}Corresponding author

Precise measurement of levels of liquids stored in tanks is essential for monitoring and predicting disasters by detecting leakages or arbitrary discharge of toxic materials. Therefore, tanks are typically equipped with a series of liquid level sensors. A magnetostrictive-type level sensor is composed of a waveguide, a current pulse interrogator, and a sensing coil for detecting reflective torsional signals caused by Wiedemann effect, which is themain principle of operation of magnetostrictive-type liquid level sensors. In order to implement a high-precision magnetostrictivetype liquid level measurement system, we used time–frequency analysis techniques such as wavelet transform (WT) to precisely detect the reflected signals. By using time–frequency analysis techniques such as short-time Fourier transform (STFT) and WT, a robust and precise liquid level measurement system can be implemented even in noisy environments.

- [1] D. S. Nyce, “Linear Position Sensors: Theory and Application,” A John Wiley & Sons, Inc., Publication, 2004.
- [2] J. P. Joule, “On a New Class of Magnetic Forces,” in the Annals of Electricity, Magnetism, and Chemistry (Conducted by William Sturgeon), Vol.8, pp. 219-224, 1842.
- [3] E. Villari, “Change of magnetization by tension and by electric current,” in Annals of Physics and Chemistry, Vol.126, pp. 87-122, 1865 (in German).
- [4] A. G. Olabi and A. Grunwald, “Design and application of magnetostrictive material,” Materials and Design, Vol.29, No.2, pp. 469-483, 2008.
- [5] F. Martínez, I. Santiago, F. Sánchez, G. Obieta, A. García-Arribas, J. M. Barandiarán, and Jon Gutiérrez, “Magnetostrictive delay line improvement for long range position detection,” Sensor and Actuators A: Physical, Vol.129, No.1-2, pp. 138-141, 2006.
- [6] E. M. Purcell, “Electricity and Magnetism,” McGraw-Hill, 1985.
- [7] A. Affanni, A. Guerra, L. Dallagiovanna, and G. Chiorboli, “Design and Characterization of Magnetostrictive Linear Displacement Sensors,” Proc. of the 21st IEEE Instrumentation and Measurement Technology Conf.(IMTC 2004), pp. 206-209, 2004.
- [8] H. Kwun and K. A. Bartels, “Magnetostrictive sensor technology and its application,” Ultrasonics, Vol.36, No.1-5, pp. 171-178, 1998.
- [9] J. Xu, X. Wu, Gongtian Shen, and Liangyun Wang, “Signal Processing for Guided Waves Testing Based on Magnetostrictive Effect,” 17th World Conf. on Nondestructive Testing, 2008.
- [10] M. Yuan, D. Sun, K. Sun, J. Wu, Z. Wang, and L. Shi, “Key Technology of Magnetostrictive Fuel Sensor,” IEEE Int. Conf. on Industrial Informatics, pp. 1000-1005, 2006.
- [11] Y. Zhang, W. Liu, H. Zhang, J. Yang, and H. Zhao, “Design and Analysis of Differential Waveguide Structure to Improve Magnetostrictive Linear Position Sensors,” Sensors, Vol.11, pp. 5508-5519, 2011.
- [12] Y. Zhang,W. Liu, J. Yang, C. Lv, and H. Zhao, “Design of Compensation Coils for EMI Suppression in Magnetostrictive Linear Position Sensors,” Sensors, Vol.12, pp. 6395-6403, 2012.
- [13] W. Zheng, C. Xiaoming, and H. Wakiwaka, “Study on Impulse Magnetic Field of Magnetostrictive Position Sensor,” Intelligent Systems and Applications(ISA), 2010 2nd Int. Workshop, Vol.II, pp. 933-936, 2010.
- [14] K. Ren, “The Influence of Temperature and Product Density to Accuracy of Magnetostrictive Level Gauge,” Proc. of 2004 IEEE Int. Conf. on Information Acquisition, pp. 248-251, 2004.
- [15] S. Yao, B. Li, Z. Wang, and X. Chang, “A Study on the Precision Improvement of the Magnetostrictive Linear Position Sensor,” Second Int. Workshop on Education Technology and Computer Science, pp. 276-279, 2010.
- [16] T. K. Worthington, P. A. Calcagno, L. T. Romankiw, and D. A. Thompson, “A Magnetoacoustic Keyboard,” IEEE Trans. on Magnetics, Vol.Mag-15, No.6, pp. 1797-1799, 1979.
- [17] H. Wakiwaka, Y. Hattori, D. Ito, T. Fujioka, K. Tashiro, and X. Chang, “Consideration of an elastic wave on a magnetostrictive linear position sensor,” IEEE Japan, LD-11-88, pp. 55-60, 2011.
- [18] M. Misiti, Y. Misiti, G. Oppenheim, and J. M. Poggi, “Wavelet Toolbox User’s Guide,” MathWorks, 2013.
- [19] M. J. Caruso, T. Bratland, C. H. Smith, and R. Schneider, “A New Perspective on Magnetic Field Sensing,” Sensors Expo Proc., pp. 195-213, 1998.
- [20] L. Kütt and J. Järvik, “Magnetic field sensor coil in fast transient measurement application,” 10th Int. Symposium : Topical Problems in the Field of Electrical and Power Engineering, pp. 117-121, 2011.
- [21] W. J. Choi and J. T. Lee, “Implementation of High Accurate Level Sensor System using Pulse Wave Type magnetostrictive sensor,” The Trans. of the Korean Institute of Electrical Engineers Vol.62, No.3, pp. 395-400, 2013.
- [22] K. Clifford and S. Fernanda: “Report: As many as 35 killed in Texas plant explosion,” The New York Times, Retrieved April 19, 2013.

*J. Adv. Comput. Intell. Intell. Inform.*, Vol.18, No.6, pp. 888-895, 2014

*J. Adv. Comput. Intell. Intell. Inform.*, Vol.18, No.6, pp. 888-895, 2014