Cutting Tool Edge and Textured Surface Measurements with a Point Autofocus Probe
Katsuhiro Miura*,†, Atsuko Nose*, Hirofumi Suzuki**, and Mutsumi Okada**
*Mitaka Kohki Co., Ltd.
1-18-8 Nozaki, Mitaka-shi, Tokyo 181-0014, Japan
**Chubu University, Aichi, Japan
To improve surface roughness, machining efficiency, and accuracy of a workpiece, measuring the roughness and contours of a cutting tool edge is crucial. However, it has not been easy for a contact stylus or non-contact methods to measure the roughness and contours of a sharp edge for two reasons: doing so damages the contact stylus and steep angles produce poor reflected rays for the non-contact method. A point autofocus probe (PAP) is widely used for the roughness and contour measurements of various precision machining surfaces. The authors have developed a new method of measuring a cutting tool edge, a method using PAP with three-axis liner stages and a rotary stage. In this study, a cutting tool edge for micro-fabrication was precisely measured, and the roughness relationships of the cutting tool edge and workpiece surface were quantitatively evaluated.
-  K. Maruno, M. Michihata, Y. Mizutani, and Y. Takaya, “Fundamental Study on Novel On-Machne Measurement Method of a Cutting Tool Edge Profile with a Flueorescent Confocal Microscopy,” Int. J. Automation Technol., Vol.10, No.1, pp. 106-113, 2016.
-  T. Asai, S. Ferdous, Y. Arai, Y. Yang, and W. Gao, “On-Machine Measurment of Tool Cutting Edge Profiles,” Int. J. Automation Technol., Vol.3, No.4, pp. 408-414, 2009.
-  ISO 25178-605:2014 Geometrical product specification (GPS) – Surface texture: Areal – Part 605: Nominal characteristics of non-contact (point autofocus probe) instruments.
-  K. Miura and A. Nose, “Point autofocus instrument,” Optical measurement of surface topography, Chapter 6, Springer, 2011.
-  R. Leach, “Point autofocus prifiling,” Fundamental principles of engneering nanometrology, pp. 139-142, 2010.
-  K. Miura, M. Okada, and J. Tamaki, “Three-Dimensional Measurement of Wheel Surface Topography with a Laser Beam Probe,” Advances in Abrasive Technology III, pp. 303-308, 2000.
-  H. Suzuki, M. Okada, Y. Masuda, Y. Namba, K. Miura, S. Morita, and Y. Yamagata, “Uitraprecision cutting of nickel plated mold for X-ray mirror,” Int. J. Automation Technol., Vol.10, No.4, pp. 624-631, 2016.
-  S. Morita, J. Guo, N. L. Yamada, N. Torikai, S. Takeda, M. Furusaka, and Y. Yamagata, “Profile measurement of a bent neutron mirror using an ultrahigh precision non-cantact measurement system with an auto focus laser probe,” Measurement Science and Technology, Vol.27, No.7, 2016.
-  S. Morita, Y. Yamagata, and J. Guo, “High speed non-contact on-machine profile measurement for ultrahigh precision machning of large free-form optical surface,” JSAT, Vol.58, No.11, pp. 705-709, 2014.
-  K. Miura, A. Nose, H. Suzuki, and M. Okada, “Development and practicality of a scanning point autofocus instrument for high speed areal surface texture measurement,” Advanced Materials Research, Vol.1017, pp. 675-680, 2014.
-  ISO 25178-605:2014 Geometrical product specification (GPS) – Surface texture: Areal – Part 605: Nominal characteristics of non-contact (point autofocus probe) instruments, 3.3.7.
-  ISO 25178-2:2012 Geometrical product specification (GPS) – Surface texture: Areal – Part 2: Terms, definitions and surface texture parameters.
-  ISO 25178-3 2012 Geometrical product specification (GPS) – Surface texture: Areal – Part 3: Specification operators.