Development of a Pneumatic Surgical Manipulator IBIS IV
Kotaro Tadano*, Kenji Kawashima*, Kazuyuki Kojima**,
and Naofumi Tanaka**
*Precision and Intelligence Laboratory, Tokyo Institute of Technology, 4259 Nagatsuta, Midori-ku, Yokohama-shi, 226-8503, Japan
**Tokyo Medical and Dental University, 1-5-45 Yushima, Bunkyo-ku, Tokyo 113-8519, Japan
In teleoperated, minimally invasive surgery systems, the measurement and conveyance of a sense of force to the operator is problematic. In order to carry out safer and more precise operations using robotic manipulators, force measurement and operator feedback are very important factors. We previously proposed a pneumatic surgical manipulator that is capable of estimating external force without the use of force sensors. However, the force estimation had a sensitivity of only 3 N because of inertia and friction effects. In this paper, we develop a new and improved model of the pneumatic surgical manipulator, IBIS IV. We evaluate its performance in terms of force estimation. The experimental results indicate that IBIS IV estimates external forces with a sensitivity of 1.0 N. We also conduct an in-vivo experiment and confirm the effectiveness and improvement of the manipulator.
and Naofumi Tanaka, “Development of a Pneumatic Surgical Manipulator IBIS IV,” J. Robot. Mechatron., Vol.22, No.2, pp. 179-188, 2010.
-  R. H. Taylor and D. Stioianovici, “Medical Robotics in Computer-Integrated Surgery,” IEEE Trans. on Robotics and Automation, Vol.19, No.5, pp. 765-780, 2003.
-  H. Yamashita, D. Kim, N. Hata, and T. Dohi, “Multi-Slider Linkage Mechanism for Endoscopic Forceps Manipulator,” In Proc. of the 2003 IEEE/RSJ, Int. Conf. on Intelligent Robots and Systems, Vol.3, pp. 2577-2582, 2003.
-  N. Matsuhira et al., “Development of a Functional Model for a Master-Slave Combined Manipulator for Laparoscopic Surgery,” Advanced Robotics, Vol.17, No.6, 2003, pp. 523-539.
-  M. Hashizume et al., “Early Experience of Endoscopic Procedures in General Surgery Assisted by a Computer-Enhanced Surgical System,” Surgical Endoscopy, Vol.16, 2002, pp. 1187-1191.
-  M. C. Cavusoglu, W. Williams, F. Tendick, and S. S. Sastry, “Robotics for Telesurgery: Second Generation Berkeley/Ucsf Laparoscopic Telesurgical Workstation and Looking Towards the Future Applications,” In Proc. of the 39th Allerton Conf. on Communication, Control and Computing, Oct. 2001.
-  F. Tajima, K. Kishi, K. Nishizawa, K. Kan, H. Ishii, M. G. Fujie, T. Dohi, K. Sudo, and S. Takamoto, “A Prototype Master-Slave System Consisting of Two MR-Compatible Manipulators with Interchangeable Surgical Tools,” In Proc. of Int. Conf. on Robotics and Automation, 2004.
-  J. Arata, M. Mitsuishi, S. Warisawa, K. Tanaka, T. Yoshizawa, and M. Hashizume, “Development of a Dexterous Minimally-Invasive Surgical System with Augmented Force Feedback Capability,” In Proc. of Int. Conf. on Intelligent Robots and Systems, pp. 3738-3743, Aug. 2005.
-  M. J. H. Lum, D. Trimble, J. Rosen, K. Fodero II, H. H. King, G. Sankaranarayanan, J. Dosher, R. Leuschke, B. M. Anderson, M. N. Sinanan, and B. Hannaford. “Multidisciplinary Approach for Developing a New Minimally Invasive Surgical Robotic System,” In Proc. of the 2006 BioRob Conf., 2006.
-  O. Gerovichev, P. Marayong, and A. M. Okamura, “The Effect of Visual and Haptic Feedback on Computer-Assisted Needle Insertion,” Computer-Aided Surgery, Vol.9, No.6, 2004, pp. 243-249.
-  J. T. Dennerlein, D. B. Martin, and H. Zak, “Haptic Force-Feedback Devices for the Office Computer: Performance and Musculoskeletal Loading Issues,” Human Factors, Vol.43, No.2, 1996, pp. 278-286.
-  A. M. Okamura, L. N. Verner, C. E. Reiley, and M. Mahvash, “Haptics for Robot-Assisted Minimally Invasive Surgery,” In Proc. of the Int. Symposium Robotics Research, Hiroshima, Japan, November 26-29, 2007.
-  G. Tholey, J. P. Desai, and A. E. Castellanos, “Force Feedback Plays a Significant Role in Minimally Invasive Surgery,” Annals of Surgery, Vol.241, No.1, 2005, pp. 102-109.
-  M. Kitagawa, D. Dokko, A. M. Okamura, and D. D. Yuh, “Effect of Sensory Substitution on Suturemanipulation Forces for Robotic Surgical Systems,” The J. of Thoracic and Cardiovascular Surgery, Vol.129, No.1, 2005, pp. 151-158.
-  S. Tachi, T. Sakaki, H. Arai, S. Nishizawa, and J. F. Pelaez-Polo, “Impedance Control of a Direct-drive Manipulator without Using Force Sensors,” Advanced Robotics, Vol.5, No.2, 1991, pp. 183-205.
-  A. J. Madhani, G. Niemeyer, and J. K. Salisbury Jr., “The Black Falcon: A Teleoperated Surgical Instrument for Minimally-Invasive Surgery,” Proc. of IEEE/RSJ Int. Conf. on Intelligent Robotic Systems, Vol.2, pp. 936-994, 1998.
-  K. Tadano, K. Kawashima, and T. Kagawa, “Evaluation of a Master Slave System with Force Sensing,” Proc. of SICE Annual Conf., pp. 2871-2874, 2007.
-  K. Tadano, K. Kawashima, and T. Kagawa, “Development of Robot Manipulator for Laparoscopic Surgery with Force Display Using Pneumatic Servo System,” Proc. of 7th JFPS Int. symposium on Fluid Power, pp. 821-824, 2008.
-  K. Tadano and K. Kawashima, “Development of 4-DOFs Forceps with Force Sensing Using Pneumatic Servo System,” Proc. of IEEE/ICRA, 2006, pp. 2250-2256.
-  Y. Yokokohji and Y. Yoshikawa, “Bilateral Control of Master-Slave Manipulators for Ideal Kinesthetic Coupling,” IEEE Trans. on Robotics and Automation, Vol.10, No.5, 1994, pp. 605-620.
-  B. Hannaford, “Design Framework for Teleoperations with Kinematic Feedback,” IEEE Tanas. on Robotics and Automation, Vol.5, pp. 426-434, 1998.
This article is published under a Creative Commons Attribution-NoDerivatives 4.0 International License.
Copyright© 2010 by Fuji Technology Press Ltd. and Japan Society of Mechanical Engineers. All right reserved.